Q:How to edit battery testing program at Neware monitor software?
- At “Brief View” toolbar,there are “Charge” ”Discharge” “Rest” "Protect" "Stop" "Pause" "Finished" "Default" items,user could click to set color for each item.
- Select the channels you want to use---right click mouse---click “single start” button into setting interface.
- Edit testing program under “standard settings”model,the first step have to be “Rest”,the time could be 2 or 3 minutes.The second step could be any charge or discharge mode,the third step suggested to be “Rest” and time could be 2 or 3 minutes too.In general,the first step have to be “Rest”,between charge and discharge step should be “Rest”.
- After step setting editing,there are three parameter need to set:”Time” under “Record Condition”,it is testing data record frequency;”Volt.Lower” and “Volt.Upper” under “Safety Limit” should set to protect tested batteries.
- Clike "OK" button to start battery testing.
Below is simple battery testing program setting picture just for reference:
Q:How to change one setting before editing battery testing programs?
- Please find the “BTS” file at the “BTSClient7.6.0” folder under Monitor Software installing road.
- Open the “BTS” file---find the “STEPCHECKMODEL” and “RECORDCHECKMODE” ---change both value into “0”.
- Save the changing.
Q:How to set mapping channels after connecting BTS to computer?
- Click “User” button and login with user name:admin and password:neware
- Right click mouse at gray zone---click “Reset Map”button---click yellow triangle(127.0.0.1) to expand---click yellow triangle(BTS82) to expand---click yellow triangle(1) to expand.
- Right click mouse at gray zone---select “The Main Channel Default Mapping” button and click “1*8”.
- Right click mouse at gray zone---click “Save Map” button.
If connect set mapping channels successfully,the monitor software interface display as below:
Q:How to install monitor software and connect BTS to computer?
- Neware BTS is controlled by the monitor software installed on the computer,user edit battery testing programs here.
- Please do not install software at C dish in case C dish have problem cause lose testing data.
- Please confirm the dish you ready to install software at have enough space.
- Change the computer IP into:192.168.1.250
- Close firewall and all antivirus software.
- Connect battery testing system to computer with network cables we offer.
- Switch on all battery testing system device.
- Install software.
If connect BTS to computer successfully,the monitor software interface display as below:
Prof Muhammad-Sadeeq Balogun of Hunan University, has nearly a decade of Chinese life experience Now he is already a real China When I first met Mr Tang, I was shocked by his fluent Chinese, and we were fortunate to be able to hear Mr Tang’s speechRead More
the first China Northwest New Energy Materials and Devices Symposium hosted by Xi an University of Technology, Advanced Electrochemical Energy Research Institute and School of Materials Science and EngineeringRead More
On December 1, 2018, the 8th International Conference on the Progress of Innovative Energy Materials (AAIEM2018), co-sponsored by Guangxi Science and Technology Association and Guangxi University, was successfully held in Guangxi UniversityRead More